Hot New Products Qs Nd Yag Laser - YAG Laser System JDS-810A – JDS
Hot New Products Qs Nd Yag Laser - YAG Laser System JDS-810A – JDS Detail:
Q-Switched Nd:YAG adopts the latest Q-switched model, makes the cure time of laser to be smallest, avoiding damage to the skin. And the skin tissue absorbs the high energy of the certain laser wavelength in instant, then expands and explodes to tiny thing easily. Then licked up by the cell and be got out of the body.The laser has two wavelengths: 1064nm&532nm. The 1064nm is for treatment of the color in blue or black pigmentation of the skin; the 532nm is for treatment of the color in red and brown pigmentation of the skin.
|Laser type||Q-switched Nd: YAG laser|
|Operating screen||8.4 dual color screen|
|Treatment head||1064nm/532nm/1320nm(Black wawa head)|
|YAG Bar Size||Phai 6|
|Aiming Beam||Guided by red light|
|Cooling system||Air Cooling & Close Water Circulation|
|Stand-by Working||Continuously stand-by working for 15 hours|
|Input power||AC 220V(110V)/ 15A 50HZ(60HZ)|
|Machine Size||450*300*340mm (L*W*H)|
The environment requirements
1. Environment temperature range: +5 ℃~ +55 ℃;
2. Relative humidity range: 30%~80%;
- Atmospheric pressure Range: 86kpa -106kpa.
Air quality：No caustic particle (such as acid) in the air, which can damage the electric wire, the electric component and optical component surfaces. The dust in the air should be as little as possible, or it can absorb the energy of light and be heated up. When the dust falls on the surface of the filter, it may damage the filter. Metal particles also could damage the electric component.
Temperature: The laser system has an optimal working temperature between 15°C~30°C. Relative humidity should not be over 80%. The working power of this device is about 2 KW. It would be best that the laser unit be installed in an air-conditioned room where the relative humidity and temperature can be maintained at optimal levels.
All persons using the equipment (including the patient) must wear the protective glasses; the eye shield for the patient and filtered 200 – 1200nm for the operators. The goggles / glasses provided with this unit are manufactured specifically for the 200nm to 1080nm wavelength produced by the laser unit . Do not substitute treatment / safety glasses with other types of tinted eyewear which may not meet the specific safety requirements of the unit.
Strictly prohibit look at the perpendicular to incidence laser and the reflected light even you are wearing protection glasses.
Moist gauze is more useful for eye protection, but the treatment part should be absolutely dry.
If the laser treatment head is leaking the system cannot switch on. If it has been switched on already, you must switch off the power immediately.
You should turn off the system to prevent fire when you are debugging the laser or cleaning the mirror of the laser.
Note: Gauze, eye-shield and tarsi are all effective in the protecting for the eyes
Product detail pictures:
Related Product Guide:
We not only will try our greatest to offer superb companies to just about every buyer, but also are ready to receive any suggestion offered by our shoppers for Hot New Products Qs Nd Yag Laser - YAG Laser System JDS-810A – JDS , The product will supply to all over the world, such as: Colombia, Madagascar, Auckland, Each product is carefully made, it will make you satisfied. Our products in the production process have got strictly monitored, because it is only to provide you the best quality, we will feel confident. High production costs but low prices for our long-term cooperation. You can have a variety choices and the value of all types are same reliable. If you have any question, do not hesitate to ask us.
By Afra from New Zealand - 2018.10.31 10:02
Staff is skilled, well-equipped, process is specification, products meet the requirements and delivery is guaranteed, a best partner!
By Merry from panama - 2017.09.29 11:19